Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I wrote a simple Labview program to read the voltage. I did it for one channel and it is measuring fine way. My purpose is to measure the voltage at two point at the same time, so I need to read the voltage at both channel at a time. Anyone can help me regarding this.
Thanks & Regards
- Keithley Applications
- Posts: 2847
- Joined: June 10th, 2010, 6:22 am
- Country: United States
So true simultaneous measurements are not possible.
One channel needs to be selected and then take a reading.
Then switch to the next channel and take another reading.
Look at the trigger model on page 7-5, figure 7-2 of the Model 2182A user manual.
Here is the link.
The Device action shows the channel in the device action.
So each reading detects the present channel.
Who is online
Users browsing this forum: No registered users and 1 guest