Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I'm trying to store a particular number of readings in the buffer and read them all at once. I've attached the program I've written to do this. I used the :TRAC:DATA? command to do this, but it doesn't appear to work.
The front panel of the instrument gets updated accordingly, but the program times out and does not give out any readings. It gives me an error saying Error -1073807339, timeout expired.
When I remove the TRAC:FEED:CONT NEXT command, there are no errors given out but it reads an empty buffer, that is, it returns all zeroes.
Why isnt this program working? Thanks in advance.
- (13.86 KiB) Downloaded 1510 times
- Keithley Applications
- Posts: 2849
- Joined: June 10th, 2010, 6:22 am
- Country: United States
2. Insert a short delay between*TRG and TRACE:DATA?.
3. TRACE:FEED:CONT NEXT is required to place data into the buffer.
That should do it.
Who is online
Users browsing this forum: No registered users and 3 guests