Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
When I read the buffer from the 2182A most of the time, but not always, the last two readings in the buffer are the exact same reading, as in they are both positive when one should be negative and not exactly the same magnitude. Somehow the second to last reading is being duplicated in the buffer. For the time being I just tell Labview to remove that point, but I'd like to know why it's happening.
I was wrong. It stopped doing it for a while, but it's right back to doing it again.
So this is still an unsolved issue.
After the buffer is read, if I take it out of trigger mode with :TRIG:SOUR IMM I get all the data points on the next sweep.
If I don't do that, on the next sweep the 2nd to last data point is duplicated. If I do it, the next sweep runs fine.
Ok, that's not it at all either.
What's the deal here?
The only thing that makes the behavior change is if I set NPLC low on both 2400 and 2182A. Or, if I set the trigger delays above a certain time. If I set NPLC levels low and trigger really fast, it won't duplicate the 2nd to last reading. If I set the NPLC of the 2182A anywhere above 2 OR the trigger delay on the 2400 above 0.03 seconds OR the trigger delay on the 2182A above 0.1 second or some combination of any of these, I get the duplicate reading problem.
Can anyone tell me what's going on here?
I believe I finally have the answer. All I had to do was put a really big number for the timeout on the RQS (polling the service request bit) in Labview. What was happening was that sweeps with slow readings were not finished until after the RQS had already timed out. If I just set the timeout to something like 60 seconds (way more than enough time for any sweep to complete) then all the points in the buffer come out fine. Before, I had just set it to something like 10 seconds and forgot about it.
Anyway, hopefully anyone else reading this will learn from my mistakes.
Who is online
Users browsing this forum: No registered users and 7 guests