Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I am running 2400 sourcemeter and 2182a nanovoltmeter into LabVIEW via Delta Mode..
I am using the Kiethley given example 2182Delta_LV6.vi and added custom varialbes for the user as well as a digital filter.
For some reason I am unable to run :SENS:VOLT:NPLC 1 or :SENS:VOLT:NPLC 5... In the user manual it recommends this setting, but when I set it to those values my RQS stage gets fully timed out (50 seconds) and my 2400 stops listening, resulting in 0s. To work I the only setting that works is 0.1.
When I "Highlight Execution" it works just fine, with either setting (1 or 5).
Why? and how can I fix this?
Adding the command
for some reason screws it up, if someone cares to explain it would be helpful.
I added it because it says it allows double Delta speed measurements in the manual.
Who is online
Users browsing this forum: No registered users and 4 guests