Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
In the first case, the AC noise will give contribution to the measured voltage. However, in the second case,the AC noise will give no contribution.
- Keithley Applications
- Posts: 2848
- Joined: June 10th, 2010, 6:22 am
- Country: United States
It is not a RMS or Root Mean Square type of measurement.
The A/D "samples" the input signal for the specific NPLC(Number of Power Line Cycles) and calculates the reading.
So from your perspective it is more of a time averaged measurement and should not be affected by the 10MHz noise.
Who is online
Users browsing this forum: No registered users and 11 guests