Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Keithley Applications
- Posts: 212
- Joined: October 25th, 2010, 1:31 pm
- Country: United States
It is not possible to connect more than one 2182A to the 6221 while using Delta mode.The Delta mode operation uses the 6221 as the master and readings are taken and displayed using the 6221 and it is designed to only communicate with only one 2182A
Who is online
Users browsing this forum: No registered users and 1 guest