Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
2502, 6485, 6487 - Support
- Posts: 1
- Joined: July 30th, 2021, 7:55 am
- Country: United Kingdom
Using the Keithley 488B GPIB to USB to connect the Keithley 6485 - to the computer. Using this setup we are able to take the measurement from the - however when connected through the GPIB, the - detects higher background noise (nA). This means we are unable to detect lower currents (pA) when connected to the Labview software. Without the GPIB connection the background noise level is around 2 pA, is there a method to fix this problem?
Who is online
Users browsing this forum: No registered users and 1 guest