Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
My apparatus is as follows: Keithley 6482, Pulse Generator, Power Supply of the Electron Gun, Electron gun itself, Sample connected to the channel of 6482.
1) Can 6482 be used as an external trigger to activate pulse generator which will the send a signal to the Power Supply of the Electron Gun and store the measured current on the buffer of 6482?
2) If this is possible, can you give me directions on how to set up the trigger and buffer to record measurements.
- Keithley Applications
- Posts: 2849
- Joined: June 10th, 2010, 6:22 am
- Country: United States
In the trigger model(page 8-3) for the front panel the ARM layer can output a pulse or the TRIG layer can output a pulse any time after the SOURCE DELAY or SENSE phase.
2. To set it up to output a pulse in the ARM layer send the following command.
This will output a pulse when it leaves the ARM layer.
To set it up to output a pulse after the SOURCE , DELAY or SENS phase, then send one of the following commands.
The pulse is a TTL level negative going longer than 10usec pulse.
Who is online
Users browsing this forum: No registered users and 1 guest