Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I have a Keithley 6487 and I need to measure currents on a high voltage line (MAX 5 kV).
I know this - has the max input voltage at about 500V, but one technician in my institute suggest me that probably it exists some external module to perform measurements at a higher voltage (up to 5 kV or more).
Does this module exist? If not, there is a way to perform this measurement with this instrument or do I have to switch to another instrument?
Thanks for your time
- Keithley Applications
- Posts: 2849
- Joined: June 10th, 2010, 6:22 am
- Country: United States
Model 6487 has its own voltage source built in. With a max output of 500V.
But if you are testing at DUT at 5kV, then make sure you do not float the 6487 at more than 500V.
Attached is a document showing a protection circuit when using high voltage with the 6487 and Keithley -.
Again, do not float the -, instead float the DUT.
Hope this helps.
Keithley technical support
Who is online
Users browsing this forum: No registered users and 5 guests