Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Using the Tek provided labview driver, we are communicating to them via dedicated ethernet. We're not doing anything fancy, just setting 5mA output.
After a while, we will lose all communication. We can't even ping. The only way to bring it back is to do a hard power cycle of the 6221.
I see people had similar issues to this in the past, and I have not seen a solution. Can anyone from Tek jump in and offer something?
Who is online
Users browsing this forum: No registered users and 1 guest