Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Keithley Applications
- Posts: 2849
- Joined: June 10th, 2010, 6:22 am
- Country: United States
On the front panel of the Model 6221 press CONFIG, then let go.
Now press DELTA.
Select I-HIGH, press ENTER.
Select the high value of current and press ENTER.
The opposite polarity of the current will be selected as the I-LOW current automatically.
Now run the Delta mode as usual.
From remote control send the following command to change the I-HIGH value to 10mA.
That should do it.
Who is online
Users browsing this forum: No registered users and 3 guests