Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
409, 41x, 42x, 44x, 48x -
14x, 18x, 1801 Nanovoltmeters
15x, 18x Null Detectors/Microvoltmeters
I've got a 182 voltmeter in which the display goes blank after a few minutes. It's losing the +60V to the display, it goes to 0V. Here are some notes:
- the +5V to the +60V VFD supply looks good even when the I lose the +60V.
- a power supply put in place of VR500 (an 8.2V zener diode) which doesn't help,
- if the display is disconnected, the +60V still stops working.
Without a schematic, I would have to guess that the +60V VFD supply is the next thing to try replacing. It is labelled: ERG Inc. E705-E1368VF. Is it possible to get one of these? Does anyone have any other suggestions on what to check?
- Keithley Applications
- Posts: 2849
- Joined: June 10th, 2010, 6:22 am
- Country: United States
There are no obsolete parts left in stock.
The Model 2182A replaces the Mode 182.
Who is online
Users browsing this forum: No registered users and 1 guest