Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Keithley Applications
- Posts: 2847
- Joined: June 10th, 2010, 6:22 am
- Country: United States
To source current between pins 1 and 2, close 1!1!1 and 1!2!2.
To measure voltage between pins 3 and 4, close 1!3!3 and 1!4!4.
To set for high resistivity mode close 1!4!5.
When in the high resistivity mode be sure to set guard jumpers on the 7065 card to "Guarded". Remove the shield and the jumpers are in the rear of the card by the triax connectors.
Also make sure to set Model 6220 and 6221 to guard mode. This is in the TRIAX menu.
Press TRIAX, select INNER-SHIELD, select GUARD.
When sourcing 100nA through 10M square sample(means 10M in each leg of square) you should see 0.25V across pins 3 and 4.
Who is online
Users browsing this forum: No registered users and 1 guest