Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I have seen that using 7001 model along with 7011-C quad is recommended for 4 probe configuration.
Could someone elaborate on this configuration and the possibilities?
I have the 7001 main frame and the 7011-C card. The last should be connected to an external micro voltmeter.
I guess you want to multiplex one current source and one voltmeter to several 4-point collinear probes.
One 7011-C card is enough to do so as you can use parts of the cards separately ('banks');
e.g. use bank A to multiplex the current source and bank B to multiplex the voltmeter between your probes.
The 7011-C card offer even further flexibility, so e.g. for van der Pauw setups
current source HI, current source LO, voltmeter HI and voltmeter LO could be all switched on individual banks.
Who is online
Users browsing this forum: No registered users and 3 guests