Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Keithley Applications
- Posts: 2849
- Joined: June 10th, 2010, 6:22 am
- Country: United States
Channel 1 could be Temperature with type T thermocouple.
Channel 2 could be a four wire RTD (Please note this would gang channel 22 for the four wire connection).
This is simple to set up in the Excelinx-1A software
Look at the DMM CONFIG sheet. Line 36.
Set channel 1 for function = Temp
Set RANGE to T
Set Option to INT
Now set channel 2 to Temp.
Set Range to PT100
That should do it.
Who is online
Users browsing this forum: No registered users and 1 guest