Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
The Tektronix PA 1000 Power Analyzer we are using has started to freeze randomly during testing. The display locks up, and no key presses seem to unfreeze it. I am unable to access the menu, and the only way to get it to work again is to turn the unit off and on again. This works for a few minutes before the screen freezes.
Who is online
Users browsing this forum: No registered users and 1 guest