Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Our organisation have one PM1000 Power Meter which was from Votech Instruments (Now taken over by Tektronics). The problem we are facing with that Power Meter is that we are getting variations in the Voltage source like if we give 230 V input voltage it shows 245 V sometimes 230 V i.e. it shows variations of +/-15 to 20V difference.
Actually in last few months we have been using the meter continuously. So is the meter sustainable to continuous use? Would the voltage variations that we are facing now might have been due to this?
Can anyone please suggest any solution to this.
Who is online
Users browsing this forum: No registered users and 2 guests