Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Here's a screenshot that shows this. The tick mark on the cursors is no where near the waveform. The value reported on the left cursor, 975 mV, is wrong too, matching the tick mark. It should be about 2.25 V. Moving the cursors does move the ticks, but they are basically always wrong. If they do line up, it would appear to just be chance.<br/>
Firmware version 1.26.0.
Is there a way to get a PNG screen shot via the web interface? While the link for the image is http://tbs2104-xxx/run/lxiprntscrn.png, with a png on the end, it's NOT a png, it's a jpg. It would look a lot cleaner with less compression noise if it was a png. I'm not interested in moving a USB drive back and forth to get screen shots. I didn't have to do that with Tek scopes from 16 years ago!
Who is online
Users browsing this forum: No registered users and 2 guests