Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I have a TDS1001B oscilloscope over 10 years old. There was a problem with USB and always checking stopped and it was necessary to press a key to follow the boot.
Now it hangs on this screen and does not proceed regardless of the key you press.
shifting and turning it on several times, it ends up working.
Has anyone ever experienced this?
Thank you for your help.
Who is online
Users browsing this forum: No registered users and 2 guests