Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Anyways... the downside is that the unit will not power up when pressing the power button.
I got a quote to repair from where we get it calibrated and it was $6,700. That is a buy another scope price! I would like to stay with the Tektronix if I can.
I have seen a reference to there being a way to do a partial teardown to get it reload the firmware. I have looked at teardown videos and there is a "CPU POWER" and a "GLOBAL RESET" button. Does anyone have experience resurrecting the MSO3000 from a failed firmware upload state?
Does anyone have experience using these buttons?
Thanks in advance.
Who is online
Users browsing this forum: No registered users and 1 guest