Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
-the probe jaw is closed
-the probe is not connected to an energized source
-the probe's internal temperature is not too high
I have verified all of these steps but so far no luck. Is there anything else I can do to try to get this probe working again?
Who is online
Users browsing this forum: No registered users and 1 guest