Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
We have AFG3102. I need to drive a non-linear circuit with changing input impedance. Is there a way to measure the inserted power to my circuit.
I need to measure power transfer efficiency. I can measure my output power but couldn't find a way to measure input power.
My circuit: a half wave rectifier
Input signal: Sine wave 4 Vp-p
Typical current less than 10uA
- Keithley Applications
- Posts: 1281
- Joined: October 15th, 2010, 10:35 am
- Country: United States
That is a challenging situation for two reasons:
- the changing input r of the load will play havoc with the afg actual applied voltage to your DUT.
- the low currents are beyond what a current probe and a scope can do for you.
You did not mention the frequencies, but consider placing a shunt resistor in series and use our DMM7510 1MHz digitizer on the voltage across the shunt. You can get your input power from the R and V.
Who is online
Users browsing this forum: No registered users and 3 guests