Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
2019-12-18 8:56:41 PM Logging Measurements
2019-12-18 8:56:41 PM Saving Data for Event 907 Complete
Exception occured during data logging.
Ivi.Visa.NativeVisaException: The connection for the given session has been lost.
at NationalInstruments.Visa.Internal.ErrorChecker.CheckStatusAndThrow(Int32 status, SafeVisaHandle visaObject)
at NationalInstruments.Visa.Internal.MessageBasedSessionImpl.Write(Byte* buffer, Int64 index, Int64 count)
at NationalInstruments.Visa.Internal.MessageBasedSessionImpl.Write(Byte buffer, Int64 index, Int64 count)
at Ivi.Visa.MessageBasedReaderWriter.FlushWrite(Boolean sendEnd)
at Ivi.Visa.MessageBasedReaderWriter.Write(Byte* pData, Int64 index, Int64 count, Boolean flush)
at Ivi.Visa.MessageBasedReaderWriter.Write(Byte data, Int64 index, Int64 count, Boolean flush)
at Ivi.Visa.MessageBasedReaderWriter.Write(Byte data, Boolean flush)
at Ivi.Visa.FormattedIO.MessageBasedFormattedIO.WriteLine(String data)
2019-12-18 8:56:49 PM Data Logger Stopped: Error
Screen capture attached. Please advise what setting is missing.
- (512.67 KiB) Downloaded 1 time
That error originates from VISA. It seem that communication with the scope was lost for some reason. I can only speculate as to why that would happen, but it has nothing to do with TekScope Utility.
I will make a note for the next release that I should add functionality to the DataLogger to try to recover from VISA errors. Right now it's one error form VISA and it quits.
Installed the tekscope utility, and see no selections available for the instrument ID area.
It is connected via usb, but there is only setup for serial....
National instruments in not installed on this machine, do we need the ni-visa installed?
is there an mso2014 driver?
We are stuck!!! HELP!
TekScope Utility requires VISA software that supports the IVI Standard VISA.NET library. This VISA software can be from any vendor as long as the vendor includes VISA.NET support. NI-VISA is the recommended VISA.
As for use on a 32-bit machine... I'm really not sure if it will work or not as I have only ever developed it for and tested on a 64-bit machine. I compiled the program using the "Any CPU" setting in the Visual Studio project and I know the .NET run-time just-in-time compiler may be able to compile it for 32-bit (x86) at run-time, but again I have never tried so I can't tell you if it will work.
Who is online
Users browsing this forum: No registered users and 1 guest