Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
According to the user manual, all I have to do to use an external GPIB Controller is select Tools---> Options---->GPIB and modify the GPIB Address (Default = 1) to something other than 1. I have tried this. When I communicate through the GPIBENET/100 Controller, the controller sees the GPIB Address of the RSA6114A (I set it to 18). But, when sending a *IDN? command (I get no errors on the send), I do not get a response from the RSA. So I though the device was set to "Listen Only". I sent the *CAL command. This should have started the Calibration Routine. Nothing happened on the RSA. So my question is, "What am I missing?" It looks lo me like the RSA is trying to be a GPIB Controller... If this is true, how do I shut it off?
You can also use Ethernet to remote control the instrument too. Does that work?
Who is online
Users browsing this forum: No registered users and 8 guests