Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Did do some further testing of the wfm. The voltages are labeled +15 -15 +5 and +40. So suspecting the switcher primary had A failure I tried provideing power to the points labeled. As I had nothing to loose anyhow And the Scope does operate. The currents apear to be ok as well so that leads to the primary area.
The switcher is produceing all the voltages just doing the cycleing thing. But i noted the 40 volt supply did go high in the cycle as much as 50 volts. I have checked all caps. And as well as all the transistors in the primary and diodes. My guess is that it leavs the opto and IC in that circut SG3524BN. As without A much needed schematic its A hard road. As just changeing parts never seams to fix things.
Also just A note the neon lamp located near the ac jack upper back of supply does oscolate. Not sure if this is normal or if this is A result of a fault.
Who is online
Users browsing this forum: No registered users and 22 guests