Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I could acquire waveform from the oscilloscope using IVI command INITIATE and FETCH, but the IVI command READ generate a timeout (see -).
Why the read command generate a Timeout?
What I missing?
- Scope_teststand.jpg (411.3 KiB) Viewed 10150 times
I can't tell from your trace what communications interface you are using with your scope, but I am going to guess it is Ethernet. This issue usually occurs because the firewall on your PC is blocking the Service Request message from the LXI protocol. To correct it you need to add an inbound exception to your firewall for the executable that is using the IVI driver, or switch your interface to USB.
I have an FAQ about this on our site at the following URL.
http://www.tek.com/support/faqs/tkdpo2k ... timing-out
Who is online
Users browsing this forum: Google [Bot] and 18 guests