Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
very useful utility, which I am using to save a hardcopy of the Screen Capture to my PC (or networked drive).
Just a thought, but it would be nice if the 'Select Instrument' panel had some 'helpful' examples of what to type in the 'VISA Resources Address' box for those that forget the format required?
For example, I was using a MSO54 via TCPIP and couldn't remember what the string format was. However, I did notice that the EScope view kindly had the string that needed to be entered. I guess the same would apply for GPIB
Thank you for the suggestion. I have added a tool-tip bubble which pops up when you hover your mouse over the VISA Resource Address field with your suggestion.
There is now a new version of the utility available for download over at the TekScope Utility thread. viewtopic.php?t=140451
it works fine with 'tekScope utility' i would like to save the screen and setup when i save the data (i can save data ok)
are you using Labview?
An as for is TekScope Utility using LabVIEW, the answer is no. It is written in C# targeting the .NET framework 4.6 and uses the generic IVI VISA.NET API for interacting with VISA.
Hope this helps!
Looks like it's been a while since anyone commented on here. We recently received a Tektronix MSO58 scope to demo (although we are looking to possibly purchase the newer 4 series version). In our test lab, we take A LOT of screen captures to verify waveforms that go into test reports. Our older Tektronix scopes all use OpenChoice Desktop to save screen captures to our network, which we use quite often and works quite well for us. We were disappointed to learn that it does not work with the newer scopes and also that there is no supported software to allow us to do this.
We were referred to your TekScope Utility. We were able to get it to work by connecting and saving a Hard Copy but unfortunately, the screen capture was difficult to view, therefore insufficient for our needs. With the Graticule set to Grid, the Grid itself was very difficult to see on the inverted (white) background. Upon changing the Graticule to Full on the scope, the Full grid was not visible at all on the inverted background. It looks like the grid colors are not inverted therefore not visible (or very difficult to see) on the inverted white background.
Is this something that can be fixed? Is it possible to add screen capture settings for how/which the grid/display window/menus show when the Hard Copy is saved? Thanks.
Just to make sure I understand this correctly, are you using the scope with the colors set to inverted? If so, are you saying that the grid is perfectly visible when you are using the scope, but in the saved image the grid is not so visible?
TekScope Utility does not manipulate the images generated by the scope so based on your description it sounds like the image the scope is producing is not very good. If you try saving a screenshot to a thumb drive on the scope itself, is the image still hard to view? The image the scope saves should look exactly like the screen looks when you are using it live. They really shouldn't differ.
"Communication with the instrument timed out while waiting for the instrument to respond to a command. Please ensure your instrument has the latest firmware installed and try again"
The software is recognizing the visa address but cannot communicate to it. Any ideas?
If you still have timeout issues, can you capture an IO Trace and attach it to your reply so I can see what command is giving it trouble?
As for the timeout issue you are having, when do you receive this timeout? What did you do in the application before the timeout occurred? What remote interface did you use? If you have an IO Trace file of the instrument communications that would help greatly.
Who is online
Users browsing this forum: No registered users and 7 guests