Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
anything I can do to correct the ch 1 undercompensation without going into the guts of the scope?
is the scope useable for hobby electronics use?
any suggestions to work around it?
- Tektronix Applications
- Posts: 137
- Joined: December 2nd, 2008, 2:27 pm
- Country: United States
(also, be sure to rotate the VAR knobs to be sure that the vertical and horizontal controls are in their calibrated position).
Who is online
Users browsing this forum: No registered users and 4 guests