Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I am new to oscilloscopes but purchased one to see if I could observe if the siginal is too weak in the new location (it’s about 250 feet from where it was originally) or if it’s getting too much interference.
I have found that the system needs to be on its own circuit and have a 2700 UH choke installed on the circuit feed but that’s about all I can come up with since apparently the manufacture has gone out of business. They do still have a website but only very limited information and no schematics.
My problem seems to be when I look at the pattern all I can identify is the 60hz siginal and don’t see any changes when energizing one of the boxes and would like to know if anyone has any suggestions for blocking out the 120V 60Hz siginal so I can see the siginal that is used to operate these boxes over the 120V line. I believe I need a high pass filter but not sure what components I would need to achieve blocking the unwanted siginal but still be able to pass the triggering siginal that the boxes use to communicate to each other.
Thanks in advance for any help you may be able to provide.
- Tektronix Applications
- Posts: 84
- Joined: December 2nd, 2008, 2:27 pm
- Country: United States
Let's assume that it operates in the 10s of kHz or higher. You certainly can start off with something like a 0.01uF (10nF) capacitor from the hot lead to a 10K resistor to Neutral, then put your scope across the 10Kohm resistor. I am assuming that you're using a battery powered scope, or at least a scope with isolated (non-grounded) inputs. And of course, use extreme caution when working with line voltages.
Now move the call box to another breaker. There is more stuff between it and the receiving boxes now. Even worse, move the call box to the other 110V circuits (the other side of the power pole transformer. Now signals have to run out to the electric utility transformer, back into the building and through a bunch of wire to the receivers.
Think about this, trace some circuits and see if that helps any.
Any system uses a signal that is transmitted over the AC connection/plug, it does that in another frequency than 50 or 60 Hz.mech11 wrote: ↑January 3rd, 2018, 9:48 pmI have a medical office call box/light system that is older probably 20 year old ...... The system uses a signal that is transmitted over the AC connection/plug.
I am new to oscilloscopes but purchased one to see if I could observe if the signal is too weak in the new location.
Now because 120V 60Hz is the signal with highest amplitude, any oscilloscope by the use of Auto-set will trigger over this signal.
Similar application this is using single phase (power line ) wires as wires for communications, this is power-line wall mount devices for Ethernet connection and internet sharing among computers in the same house.
Who is online
Users browsing this forum: No registered users and 10 guests