Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I set the V-Limit of Keithley 220 as 100V, but it cannot go past the current at which the voltage hits around 25V. Is there any other setup that might put a cap in voltage before the V-Limit we set on the front panel of 220? I'm running it via LabVIEW programs.
- Keithley Applications
- Posts: 402
- Joined: June 10th, 2013, 6:00 am
- Country: United States
Make sure that you use an adapter that drops guard and does not short guard to LO/SHIELD.
You must use an appropriate triax adapter at the rear of the instrument. Using a BNC adapter will likely ruin the triax jack on the rear of the instrument.
Who is online
Users browsing this forum: No registered users and 20 guests