Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Posts: 1
- Joined: August 17th, 2018, 7:11 am
- Country: United States
On many of the TDS scopes the message ' Wfm does not cross ref' appears for C1, C2, or both. These scopes may or may not pass some or all of the start up and calibrations tests. I cannot find the meaning or significance of this message in the service manuals. Can anyone explain the meaning of this screen message? Thanks!
- Posts: 17
- Joined: September 1st, 2018, 11:27 am
- Country: Greece
I suppose that Wfm this is the sort version of the word : Waveform.
Therefore the message could be that : Waveform this does not cross reference points ... possibly because signal amplitude this is not enough or higher than what it is required for calibration.
Who is online
Users browsing this forum: No registered users and 21 guests