Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I cannot find in Programmer Manual how to press this button programmatically.
Please suggest what command should I use.
- Tektronix Applications
- Posts: 78
- Joined: December 2nd, 2008, 2:27 pm
- Country: United States
This will setup the Single acquisition.
Then, ACQUIRE:STATE ON will put the scope in Run mode, waiting for the trigger event.
Once the scope triggers, it will be stopped.
To re-arm the Single mode, issue ACQUIRE:STATE ON again...
Who is online
Users browsing this forum: No registered users and 1 guest