Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I have an old 465 M scope that I bought a while back. Recently I started using it and it worked without any hitch until one night I forgot to switch it off. Now there is no trace and the fuse F558 keeps blowing: Figure FO-9 CRT and Z axis schematic. I have checked the nearby components -in circuit- and do not see anything that has failed. Could it be the transformer T550 failed due to heat build up? How could it be checked and is there a spare parts supplier for these?
For an old scope of this age, it works beautifully and I hate to junk it as it serves my limited work on old equipment and would not justify spending on a new/expensive unit at a time I cannot afford while retired!
Thanks for any input anyone can provide.
Who is online
Users browsing this forum: No registered users and 4 guests