Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Posts: 1
- Joined: June 29th, 2020, 10:20 am
- Country: Canada
I am using the keithley 6485 driver on labview. I am having a hard time keeping zero check and zero correct on. Here is the code I am using. I turn zero check on but it turns off as soon as measurements are triggered. I am having a similar problem with zero correct. Ideally, I want to turn on zero check, turn on zero correct, and then turn off zero check before measuring the current.
Who is online
Users browsing this forum: No registered users and 3 guests