Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Keithley Applications
- Posts: 2841
- Joined: June 10th, 2010, 6:22 am
- Country: United States
1. Connect the 10MHz output on one unit to the 10MHz input on the other unit(s).
2. Set all instruments to the same frequency.
3. Set the phase relationship between the instruments by
pressing UTILITY and selecting OUTPUT SET UP.
4. Then select Adj Phase.
Adjust the phase until you get the required phase between the units.
Use an oscilloscope to verify the phase relationship
Who is online
Users browsing this forum: No registered users and 1 guest