All I want the load to do is turn off if the voltage drops below a set value. Is there a way to do this from the front panel? The VON setting doesn't seem to do anything.
- Keithley Applications
- Posts: 1615
- Joined: October 15th, 2010, 10:35 am
- Country: United States
If the required duration is longer than the 99999 seconds supported by the 2380 load, a SourceMeter such as model 2450 may serve you better. With the built-in scripting in the instrument, you could easily implement a loop until the measured V falls to XX level while sinking a prescribed amount of current to discharge a battery. It could even save the V vs. Time model to a thumb drive, if desired.
Who is online
Users browsing this forum: No registered users and 1 guest