Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
DMM6500 6½-Digit Touchscreen Multimeter
- Posts: 1
- Joined: October 19th, 2021, 12:57 pm
- Country: United States
Is there a VI or another way to set up the trigger block, relatively simply, so that I can read mulitple channels off a 7708 switch? This was easy to do with the 2700 and is easy to read with single samples, but multiple samples give me a trigger block error. This is frustrating.
Who is online
Users browsing this forum: No registered users and 2 guests