Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
1. Download the attached .tsp file
2. Make sure your DMM’s command set is set to TSP in MENU > Settings
3. Put the script on a USB drive and insert into the DMM
4. Press MENU then touch "Run" in the "Scripts" column
5. Select the script named "MeasurePower.tsp"
6. Click "Run Selected"
The script leverages the fact that the Ratio function stores a voltage measurement for both Sense and Input voltages in a single reading, displaying Input voltage over Sense voltage. If we use a shunt resistor across the Sense terminals, we can measure current because we know the shunt resistor value and we measure the voltage across the resistor. The Sense voltage value is stored in the extra field of the buffer, so we make the reading buffer a "Full" style buffer to include this value. The shunt resistor value is divided out of the Sense voltage measurement according to Ohm's law, giving us a value for current. The Sense voltage is also multiplied out of the Ratio measurement to get Input voltage. The two are then multiplied together to find power and the result is stored in a writable buffer, which is shown on the instrument's screen. The timing and number of measurements can be adjusted by changing the "timeBetween" and "readings" variables in the script.
- (1.16 KiB) Downloaded 253 times
Who is online
Users browsing this forum: No registered users and 1 guest