Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Keithley Applications
- Posts: 1436
- Joined: October 15th, 2010, 10:35 am
- Country: United States
I tested both fixed 100mA range and auto range on the DMM.
The SMU was fixed ranges for both sourcing and measuring.
Simple setup sketch: Outcomes:
Who is online
Users browsing this forum: No registered users and 1 guest