Scanning Card: 7700
I try to use pc to scan and read, the command and procedure is OK.
I try to use PC to read the scanning results when TSP Script is under running. But failed.
Who can know how to do it?
- Keithley Applications
- Posts: 1601
- Joined: October 15th, 2010, 10:35 am
- Country: United States
And in general, what is the use case? Is it a very long duration scan and you just want data during it?
The TSP script that runs on the instrument should avoid use of waitcomplete() as new commands sent from PC will not be processed during the wait.
You’ll most likely need to use the defbuffer1.n value to know the how much data is available.
Who is online
Users browsing this forum: No registered users and 1 guest