Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Keithley Applications
- Posts: 1615
- Joined: October 15th, 2010, 10:35 am
- Country: United States
If scan not used, then one by one you can close channel and do a dmm measure. But you must send result to PC or store it somehow before next reading on next channel.
I think on another of your posts, you are using DAQ6510 from VEE. What is the difficulty with using scan and obtaining the data from buffer with the trace data command?
What bothers me when I do loops with 1 scan is that the display of the device constantly changes from one of the values read, to a display with dashes (- - -, - -).
I also cannot use the device graph which is initialized at each scan
Who is online
Users browsing this forum: No registered users and 1 guest