Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I am currently working on a proyect with 2700 DMM with 7700 multiplexer card placed at slot 1(20 channels), connecting with my computer on GPIB and programming on LabView. In this proyect I need to measure 10 channels on Voltage and other 5 on Resistance but my DMM does not change mode (from voltage to resistance). My commands on LabView are:
SENS:FUNC 'VOLT:DC' , (@101:110)
VOLT:DC:RANG 1 , (@101:110)
SENS:FUNC 'RES' , (@111:115)
RES:RANG 1e3 , (@111:115)
Is there any of the programming wrong or something missing?
Who is online
Users browsing this forum: No registered users and 5 guests