Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I have bought 2700 DMM and found that the slots (rare panel) for channel input were empty. I would like to use it for measuring the out put of a pressure transducer, resistance of a heater simultaneously. Please tell me what I am supposed to do?
If you want to use the rear panel of the 2700 to connect to multiple DUTs you would need to purchase a plug-in module such at the 7700 which would allow you to measure 20 different devices.
For the transducer you would be able to reach the voltage of it and you could apply a math function to get the resulting pressure and you would be able to read the resistance of the heater, however you will not be able to do both of these measurements simultaneously you would only be able to do the measurements sequentially.
- Keithley Applications
- Posts: 1280
- Joined: October 15th, 2010, 10:35 am
- Country: United States
The 32 channel 7703 does not have screw terminals. You'd have to connect to the 50-pin D-sub connectors.
A technical data book that highlights each of the available card types for the 2700 Series can be downloaded from here:
Who is online
Users browsing this forum: Google [Bot] and 4 guests