Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I am trying to access 3706A web interface page but not able to load the IP address on my browser(using Chrome browser).
I checked the IP address,gateway and subnet mask properly, 3706A system is connected to the PC through LAN crossover cable.
HTTP Error 404. The requested resource is not found.
- Keithley Applications
- Posts: 1244
- Joined: October 15th, 2010, 10:35 am
- Country: United States
If no ping, it is address or subnet issue mismatch between computer and instrument.
Who is online
Users browsing this forum: No registered users and 1 guest