Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Web interface has no way to save buffer data. Only thing that's possible is to export graphical file of graphed data (export chart image).
Tried manually on front panel and I get an error 2203 "cannot open file". Tried multiple USB drives with different formats (2gig, 8gig, 16gig).
Why is there no way to get the raw data from the web interface and why so hard to get it on USB? Software updates anyone?
Also, this interface doesn't like the word "P" "I" "C" ... when you try to preview the post it turns it into a dash
- Keithley Applications
- Posts: 402
- Joined: June 10th, 2013, 6:00 am
- Country: United States
If you want to save data to a PC directly, you have to use Labview, C++, Python, etc. Your choice for user interface. I have successfully used Python with a customer to create a remote data logging measurement system.
3706A with firmware 1.53c
1 - 3722 Dual 1X48 Multiplexer card.
Connected to laptop through ethernet cable (crossover)
I thought I could set this instrument up quickly to do simple scan of 5 channels (voltage measurement) every 30 seconds (200 total measurements) without having to do a program. I can do it through the web interface (scan builder) but there is no way to get the raw data that I can see (no button to dump buffer to USB stick). I can get a graph of the data only.
So, this instrument is not designed to do this type of quick measurement?
Also tried to use the test script builder to do it, but that won't recognize the 3706 is even connected through the instrument console.
I just want to get the raw data dumped to usb stick after the scan is complete.
..... where testdata is the buffer name and mydata is the .csv file on the usb drive
Who is online
Users browsing this forum: No registered users and 5 guests