Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I choose ' 4x Prees whell' I change 'no function' with wheel. For example, I choose 'DC Volts' press wheel after l saw 'Error 1114' . How to solve ? Another error is ' Error 1116 ' I don't understand this error, why ?
Another problem control card 3740 . I wanna measure 2-wire ohm but 'overflow'. You can see below -. Where did l do wrong ? How to solve I ?
Thanks for helpful.
- WhatsApp Image 2017-07-19 at 11.45.46.jpeg (250.37 KiB) Viewed 7244 times
- WhatsApp Image 2017-07-19 at 11.45.45.jpeg (177.5 KiB) Viewed 7244 times
- WhatsApp Image 2017-07-19 at 11.45.44.jpeg (308.7 KiB) Viewed 7244 times
- Keithley Applications
- Posts: 1280
- Joined: October 15th, 2010, 10:35 am
- Country: United States
They do not connect external signals to the multimeter within the 3706A.
This is why your attempts to associate a measure function with a 3740 channel is resulting in various errors about invalid configuration.
http://www.tek.com/keithley-switching-a ... anual/seri
Who is online
Users browsing this forum: No registered users and 1 guest