Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
2302, 2303, 2303-PJ, 2304, 2306, 2306-PJ, 2306-VS, 2308
- Posts: 1
- Joined: June 27th, 2018, 10:00 pm
- Country: Taiwan
My problem is that the current line of the digital display is reading OVERFLOW. When the instrument wasn't wired to anything I was getting a current reading that jumped between 1-2 A so I wired a 10 ohm, 5 W resistor and set the voltage to 1V. I assumed this would give me a current of .1A (.1A *10ohm = 1V) with a power drain of 0.1W (0.1A^2 * 10ohm = 0.01 *10 = 0.1W).
The current line has been reading OVERFLOW no matter what I do. The current limit is set to 5A and, though I am certain this is a display issue rather than a screwup in the circuit (since the DMM is reading correctly).
Who is online
Users browsing this forum: No registered users and 1 guest