Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
2302, 2303, 2303-PJ, 2304, 2306, 2306-PJ, 2306-VS, 2308
- Posts: 1
- Joined: December 5th, 2011, 7:01 am
- Country: Brazil
I am trying get currentvaleu by GPIB , and I wanna know which command(procedure) I send to get the current value that show in display
- Keithley Applications
- Posts: 254
- Joined: February 28th, 2011, 6:17 am
- Country: United States
FETCH? will return the last reading. READ? will initiate another reading to be taken and fetch the reading back to your computer. Both of those commands however require you to set up your measurement first as you would like to do it. Typically that is done in the SENSE subsystem portion of the commands. The 2304A user manual has a section in the SCPI command reference (section 4) showing you all of the commands possible with that and section 3 goes over all of the GPIB operation. Below is the link to that manual.
Who is online
Users browsing this forum: No registered users and 1 guest