Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Keithley Applications
- Posts: 1323
- Joined: October 15th, 2010, 10:35 am
- Country: United States
The model 2308 measures pulsed currents with the following features:
- measure during the high portion of pulse
- measure idle current or low portion of pulse
- measure the average current
It uses rising edge above a programmable threshold to detect the high pulse and uses falling edge detection for the low portion of pulse.
The integration time for the measurement is programmable in increments of 33.3usec.
The pulse widths can be as short as 50usec and as long as 833msec.
Here is a link to the product page:
http://www.keithley.com/products/dcac/h ... y/?mn=2308
Who is online
Users browsing this forum: No registered users and 5 guests