I am using Keithley 224 programmable current source.
How to send External trigger command to any other instrument form 224 Current Source?
- Keithley Applications
- Posts: 402
- Joined: June 10th, 2013, 6:00 am
- Country: United States
Please see the model 224 Quick Reference Guide. On page 18, it states that the negative going trigger out pulse happens at the end of a dwell time. The 224 Instruction Manual page 2-15 states that the external trigger output is present only when a source value has completed the increment or decrement.
If you want to programmatically send a trigger, you may consider using the digital out signals.
Who is online
Users browsing this forum: No registered users and 1 guest